Minimum system requirements for RangeVision NEO and Spectrum:
RangeVision NEO scans objects between 3 cm and 1.2 meters.
RangeVision Spectrum - from 10 mm to 3 meters.
RangeVision PRO - from 5 mm to 5 meters.
The resolution of the devices will not be sufficient to scan smaller objects. When scanning objects larger than the maximum allowed size, the quality of the result will be negatively affected by the accumulated error and a lot of data.
The projector light shouldn't be absorbed. Highly reflective, translucent or very dark surfaces must be covered with matt spray.
Scanning area is scanner's field of view. RangeVision scanners have several scanning areas. You need to use different scanning zones for objects of different size.
You can scan with or without markers or using the turntable. Scans can be aligned automatically or manually.
Set of calibration plates;
Set of markers;
Set of cables;
ScanCenter NG software ;
Automatic rotary table;
Protective travel case;
Licence key (except NEO).
RV PRO standard set also includes replaceable optics for each scanning zone and a control computer (without I / O devices).
You can find information about resolution and accuracy of 3D scanner technical specification. These attributes depend on scanner model and scanning area. Specified values can be achieved by following calibration rules.
Calibration must be performed before the first scan and after changing a scan zone.
The scanner can be de-calibrated under the influence of the heating and cooling cycles of cameras and a projector, as well as when you move it. Therefore, we recommend you perform a full calibration at the beginning of each working day and after the scanner has been turned off for a long time (> 2 hours).